system refurbishment procedures. All equipment to be in 100% operating condition.
Replace ALL Treatment Table bearings
b. Provide NEW sub-base for system installation
REBUILT Pneumatic source system, solenoids, air cylinder, hoses, etc.
Replace all belts
Replace all light bulbs.
Inspect all gantry and source head bearings – replace all that do
not meet original manufacturers performance specifications.
Regrease all bearing assemblies.
New string on Treatment Table ‘tennis raquet”.
All cables to be inspected for
continuity and replaced as necessary
All relay contacts to be inspected and cleaned or replaced as necessary.
All motors to be tested for 100%
voltage and torque and replaced as necessary.
All solder contacts to be inspected
and replaced as necessary
New source drawer ‘rider rings’ to be installed
NEW source cable return spring unit
Tabletop is replaced with
the FTS Carbon Fiber Patient Table.
All exterior metal is sanded,
primed and repainted.
All cracks in ABS covers are
professionally repaired and repainted to AS-NEW condition.